Jonathan Mark Fazekas - Hebron KY, US
Cincinnati Test Systems, Inc. - Cleves OH
A fill time and stabilize and measurement time are determined for a pneumatic testing procedure. The fill time is computed by identifying, in the filling of a sample unit, a time when a substantial portion of variations of a measured variable (due to stabilization) are completed. The stabilize and measurement time is determined by comparison of the measurement variable behavior when filling sample unit(s) that do not leak, to the measurement variable behaviors when filling a sample unit that is in communication with an orifice simulating a leak. A measurement performance factor may be computed from this data at each of several possible times after a unit is filled. A gauge repeatability factor may be computed based on variance of nonleaking units at each possible time, compared to the variable change made on a sample unit when coupled to a leak-simulating orifice. Either factor may be used to select a stabilize and measurement time.